—— Mr. David Chambers
—— Neil Renato
—— Fermin Lee
—— Dalius Skinulis
—— Sun Chull Kim
—— Sabinian Smith
How Drop Impact Test Machine Works?
Drop Impact Test machine is to test large and heavy package products’ drop impact performance, through simulates real face ,edge and corner drop conditions.
This equipment mainly used to inspect package products drop resistance capability during transportation ,load and unload works, thus help improve the package design.
1. Fix the specimen on the drop arm, lift to the certain drop height, only±2% difference allowed between the height and setting value. Drop height refers to the distance between the lowest point when release the specimen and the shock impact plate.
2. Release the specimen according to below predetermined state:
A. Face dropping, the angle between the specimen drop face and flat plane should within 2º;
B. Edge dropping, the angle between the edge and the flat plane within 2º,the angle between upper face of specimen and shock plate within ±5ºor 10% of angle(Whichever is the larger value),ensure the specimen gravity line through the dropping edge.
C. Corner dropping, the angle tolerance between the specimen upper face and shock impact plate less than or 10% of the angle(Whichever is the larger value),ensure the specimen gravity line through the dropping corner..,
D. No matter what kind shape or status of the specimen, all have to ensure the gravity line of the specimen through the dropping face, edge and corner.
3. The difference between the actual shock speed and the speed when free dropping should within ±1% .
4. After testing, check the package and inner products damage condition according to related standard or regulations. And analyze the testing result.
|Factory Address:No. 88 Jiaolian Yiheng Rd, Wanjiang District, Dongguan, China, 523046|
|Sales office:Room 301, Building No.1, No. 2 Jiaolian Yiheng Rd, Wanjiang District, Dongguan, China, 523046|