Payment & Shipping Terms:
|Height:||Max Height Of Specimen 400mm||Loading:||Motorized Loading Control|
|Reading:||Hardness Shows On LCD Screen||Data Output:||By RS232, Built-in Mini Printer|
|Resolution:||0.1HR||Dwell Time:||0 - 60 Seconds, Adjustable|
Digital Rockwell Hardness Testing Machine,
Full Scales Rockwell Hardness Testing Machine,
Digital Rockwell Hardness Tester
Max Height 400mm Digital Full Scales Rockwell Hardness Testing Machine with Built-in Printer
iRock-432DL Digital Rockwell hardness tester equips with a newly-designed large displaying screen with good reliability, excellent operation and easy watching.
It can show and set the present scale, test force, test indenter, dwell time and hardness conversion.
The main functions as follows:
1, Support all the Rockwell scales;
2, Support Hardness Conversion;
3, Test results can be saved and printed;
4, Automatically calculate the maximum, minimum and average value;
5, With RS232 interface for data transferring to computer.
Suitable for quenched steel, tempered steel, annealed steel, cold and hard casting, malleable cast iron, hard alloy steel, aluminum alloy, copper alloy, bearing steel etc.
|Scale Symbol||Indenter||Test Force (Kgf)||Dial Figures||Typical Application of Scales|
|HRB||1/16” Ball||100||red||Copper alloys, soft sheets, aluminum alloys, malleable iron, etc.|
|HRC||Diamond||150||black||Steel, hard cast irons, pearlitic malleable iron, titanium, deep case hardened steel, and other materials harder than B100.|
|HRA||Diamond||60||black||Cemented carbides, thin steel, and shallow case-hardened steel.|
|HRD||Diamond||100||black||Thin steel and medium case hardened steel, and pearlitic malleable iron.|
|HRE||1/8” Ball||100||red||Cast iron, aluminum and magnesium alloys, bearing metals.|
|HRF||1/16” Ball||60||red||Annealed copper alloys, thin soft sheet metals.|
|HRG||1/16” Ball||150||red||Malleable irons, copper-nickel-zinc and cupro-nickel alloys. Upper limit G92 to avoid possible flattening of ball.|
|HRH||1/8” Ball||60||red||Aluminum, zinc, lead.|
|HRK||1/8” Ball||150||red||Bearing metals and other very soft or thin materials. Use smallest ball and heaviest load that does not give anvil effect.|
|Initial Test Force||10kgf (98.07N)|
|Total Test Force||60kgf (558.4N), 100kgf (980.7N), 150kgf (1471N)|
|Indenter||Diamond Rockwell Indenter, ф1.588mmBall Indenter|
|Loading Method||Automatic (Loading/Dwell/Unloading)|
|Hardness Reading||LCD Screen Display|
|Test Scale||HRA, HRD, HRC, HRF, HRB, HRG, HRH, HRE, HRK, HRL, HRM, HRP, HRR, HRS, HRV|
|Conversion Scale||HV, HK, HRA, HRB, HRC, HRD, HRE, HRF, HRG, HRK, HR15N, HR30N, HR45N, HR15T, HR30T, HR45T, HS, HBW|
|Data Output||Built-in Printer,RS232 Interface|
|Max. Height of Specimen||400mm|
|Power Supply||AC220V/50Hz/1Ph, AC110V/60Hz/1Ph|
|Execute Standard||ISO 6508, ASTM E-18, JIS Z2245, GB/T 230.2|
|Dimension||520×215×880mm, Packing Dimension: 650×370×1150mm|
|Weight||Net Weight: 98kg, Gross Weight: 120kg|
|Instrument Main Body||1 set||Diamond Rockwell Indenter||1 pc|
|ф1.588mmBall Indenter||1 pc||Middle Plane Test Table||1 pc|
|Large Plane Test Table||1 pc||V-shaped Test Table||1 pc|
|Hardness Block 60~70 HRC||1 pc||Hardness Block 20~30 HRC||1 pc|
|Hardness Block 80~100 HRB||1 pc||Weight A, B, C||3 pcs|
|Power Cable||1 pc||Fuse 2A||2 pcs|
|Level||1 pc||Horizontal Regulating Screw||4 pcs|
|Screw Driver||1 pc||Spanner||1 pc|
|Anti-dust Cover||1 pc||Usage Instruction Manual||1 copy|
Principle of Rockwell Test:
The Rockwell test consists of measuring the additional depth to which a carbide ball or diamond penetrator is forced by a heavy (major) load beyond the depth of a previously applied light (minor) load (SET point).
The minor load is applied first and a SET position is established on the dial gauge or displacement sensor of the Rockwell tester. Then the major load is applied. Without moving the piece being tested, the major load is removed and, with the minor load still applied, the Rockwell hardness number is automatically indicated on the dial gauge or digital display.
The diamond penetrator is used for testing materials such as hardened steels and cemented carbides. The carbide ball penetrators, available with 1/16 inch, 1/8 inch, 1/4 inch, and 1/2 inch diameter, are used when testing materials such as steel-copper alloys, aluminum and plastics to name a few.
Rockwell testing falls into two categories: Regular Rockwell testing (e.g., C and B scales) and Rockwell superficial testing (e.g., 30 N and 30 T scales).
High Rockwell hardness numbers represent hard materials and low numbers soft materials.
Contact Person: iqualitrol
CCD vision automatic measurement system: including image capture, calibration, image processing, geometric measurement, document annotation, album management, and fixed-scale printing
Automatic measurement: automatically capture the indentation and measure the indentation diameter and calculate the corresponding Brinell and Vickers hardness values
Max Specimen Height: Rockwell 175mm, Brinell/Rockwell 115mm
Automatic calibration: full image calibration on the intersection points of the calibration grid