Payment & Shipping Terms:
|Test:||Mode HV HK||Turret:||Automatic Turret|
|Objective:||10x, 40x||Light:||Halogen Lamp|
|Vertical:||Max Height 185mm||Horizontal:||Max Throat 130mm|
Micro Digital Hardness Testing Machine,
Digital Vickers Hardness Tester
CCD System Micro Digital Hardness Testing Machine Automatically Turning Turret
Technical Data Download:iqualitrol Micro Hardness Tester iVick-458ST
iVick-458ST is a new generation of Micro Vickers hardness tester adopts the integrated design of hardness tester and panel computer, all the testing parameters can be selected on the panel computer.
Touch screen makes the operation quickly and conveniently.
CCD image acquisition system can show dynamic indentation image, lock the image and automatically get the Vickers hardness value with high precision and stability, and avoid the human errors.
iVick-458ST adopts the integrated design of hardness tester and panel computer, with Windows 7 operating system, it has all functions of computer and can be connected to the output devices such as external screen and printer;
iVick-458ST equips with motorized turret between the objective and the indenter;
The lifting screw of iVick-458ST adopts worm and gear structure with smooth transmission;
iVick-458ST comes with the function of hardness scale conversion;
Automatically save the measuring data, generate the hardness-depth curve and save as WORD or EXCEL report;
Digital micrometer X-Y test table makes measurement more precise and realizes the fully automation of Micro Vickers hardness testing with optional motorized X-Y test table.
Suitable for ferrous metal, non-ferrous metals, IC thin sections, coatings, ply-metals; glass, ceramics, agate, precious stones, thin plastic sections etc.; hardness testing such as that on the depth and the trapezium of the carbonized layers and quench hardened layers.
|Test Force||gf||10gf, 25gf, 50gf, 100gf, 200gf, 300gf, 500gf, 1000gf|
|N||0.098N, 0.246N, 0.49N, 0.98N, 1.96N, 2.94N, 4.90N, 9.80N|
|Loading Method||Automatic (Loading/Dwell/Unloading)|
|Computer||CPU: Intel I3, Memory: 2G, SSD: 32G|
|CCD Pixel||1.30 Million|
|Conversion Scale||HV, HK, HRA, HRB, HRC, HRD, HRE, HRF, HRG, HRK, HR15N, HR30N, HR45N, HR15T, HR30T, HR45T, HS, HBW|
|Hardness Reading||Indentation Displaying and Automatic Measuring on Touch Screen|
|Data Output||WORD or EXCEL Report with Curve Chart|
|Objective||10x (Observe), 40x (Measure)|
|Total Magnification||100x, 400x|
|Light Source||Halogen Lamp|
|Motorized X-Y Test Table||Size: 100×100mm; Travel: 25×25mm; Resolution: 0.01mm|
|Max. Height of Specimen||185mm|
|Power Supply||AC220V/50Hz/1Ph, AC110V/60Hz/1Ph|
|Execute Standard||ISO 6507, ASTM E384, JIS Z2244, GB/T 4340.2|
|Dimension||560×335×675mm, Packing Dimension: 650×380×960mm|
|Weight||Net Weight: 48kg, Gross Weight: 62kg|
|Instrument Main Body||1 set||Diamond Micro Vickers Indenter||1 pc|
|10x, 40x Objective||each 1 pc||Weights||6 pcs|
|Weight Axis||1 pc||Cross Test Table||1 pc|
|Flat Clamping Test Table||1 pc||Thin Specimen Test Table||1 pc|
|Filament Clamping Test Table||1 pc||Horizontal Regulating Screw||4 pcs|
|Level||1 pc||Power Cable||1 pc|
|Fuse 1A||2 pcs||Halogen Lamp 12V/15~20W||1 pc|
|Screw Driver||2 pc||Hardness Block 400~500 HV0.2||1 pc|
|Hardness Block 700~800 HV1||1 pc||Anti-dust Cover||1 pc|
|Usage Instruction Manual||1 copy|
The Knoop hardness test method is mostly used for small parts, thin sections, or case depth work. Knoop indentation is very small, the main application like below:
testing very thin materials like foils or measuring the surface of a part,
small parts or small areas,
measuring individual microstructures,
measuring the depth of case hardening by sectioning a part and making a series of indentations to describe a profile of the change in hardness.
The Knoop indenter is a pyramid shaped diamond is more elongated or rectangular in shape. The Knoop method is commonly used when indentations are closely spaced or very near the edge of the sample.
The width of the Knoop indentation can provide more resolution for measurement and the indentation is also less deep. Consequently, it can be used on very thin materials.
The Vickers hardness test method consists of indenting the test material with a diamond indenter, in the form of a right pyramid with a square base and an angle of 136 degrees between opposite faces subjected to a load of 1 to 100 kgf. The full load is normally applied for 10 to 15 seconds.
The two diagonals of the indentation left in the surface of the material after removal of the load are measured using a microscope and their average calculated. The area of the sloping surface of the indentation is calculated. The Vickers hardness is the quotient obtained by dividing the kgf load by the square mm area of indentation.
When the mean diagonal of the indentation has been determined the Vickers hardness may be calculated from the formula, but is more convenient to use conversion tables.
The Vickers hardness should be reported like 800 HV/10, which means a Vickers hardness of 800, was obtained using a 10kgf force. Several different loading settings give practically identical hardness numbers on uniform material, which is much better than the arbitrary changing of scale with the other hardness testing methods.
The advantages of the Vickers hardness test are that extremely accurate readings can be taken, and just one type of indenter is used for all types of metals and surface treatments. Although thoroughly adaptable and very precise for testing the softest and hardest of materials, under varying loads, the Vickers machine is a floor standing unit that is more expensive than the Brinell or Rockwell machines.
Contact Person: Mr. Raymond Chung