Payment & Shipping Terms:
|Loading:||Motorized Loading||Hardness:||Dial Reading 0.5HR|
|Dwell Time:||0 - 60 Seconds, Adjustable||Initial Force:||10Kgf (98N)|
rockwell hardness test equipment,
hardness testing instruments
Dial Gauge 0.5HR Motorized Loading Regular Rockwell Hardness Tester Throat 165mm
Technical Data Sheet: iqualitrol Hardness Tester VEXUS SHR-150E
Operation Manual: iqualitrol VEXUS SHR-150E Operation Manual
iqualitrol Vexus SHR-150E Motorized Rockwell hardness tester adopts the mechanism of automatic loading and unloading of the test force, the test force is regulated by the load-change hand wheel.
During operation iqualitrol Vexus SHR-150E, the users can select the dwell time. Besides setting zero to dial, there is no man-made error. The instrument has high sensitivity and stability.
Suitable for quenched steel, tempered steel, annealed steel, cold and hard casting, malleable cast iron, hard alloy steel, aluminum alloy, copper alloy, bearing steel etc.
|Initial Test Force||10kgf (98.07N)|
|Total Test Force||60kgf (558.4N), 100kgf (980.7N), 150kgf (1471N)|
|Indenter||Diamond Rockwell Indenter, ф1.588mmBall Indenter|
|Loading Method||Automatic (Loading/Dwell/Unloading)|
|Max. Height of Specimen||175mm|
|Power Supply||AC220V/50Hz/1Ph, AC110V/60Hz/1Ph|
|Execute Standard||ISO 6508, ASTM E-18, JIS Z2245, GB/T 230.2|
|Dimension||520×215×700mm, Packing Dimension: 650×370×950mm|
|Weight||Net Weight: 78kg, Gross Weight: 100kg|
|Instrument Main Body||1 set||Diamond Rockwell Indenter||1 pc|
|ф1.588mmBall Indenter||1 pc||Middle Plane Test Table||1 pc|
|Large Plane Test Table||1 pc||V-shaped Test Table||1 pc|
|Hardness Block 60~70 HRC||1 pc||Hardness Block 20~30 HRC||1 pc|
|Hardness Block 80~100 HRB||1 pc||Weight A, B, C||3 pcs|
|Power Cable||1 pc||Fuse 2A||2 pcs|
|Level||1 pc||Horizontal Regulating Screw||4 pcs|
|Screw Driver||1 pc||Spanner||1 pc|
|Anti-dust Cover||1 pc||Usage Instruction Manual||1 copy|
The Rockwell test consists of measuring the additional depth to which a carbide ball or diamond penetrator is forced by a heavy (major) load beyond the depth of a previously applied light (minor) load (SET point).
The minor load is applied first and a SET position is established on the dial gauge or displacement sensor of the Rockwell tester. Then the major load is applied. Without moving the piece being tested, the major load is removed and, with the minor load still applied, the Rockwell hardness number is automatically indicated on the dial gauge or digital display.
The diamond penetrator is used for testing materials such as hardened steels and cemented carbides. The carbide ball penetrators, available with 1/16 inch, 1/8 inch, 1/4 inch, and 1/2 inch diameter, are used when testing materials such as steel-copper alloys, aluminum and plastics to name a few.
Rockwell testing falls into two categories: Regular Rockwell testing (e.g., C and B scales) and Rockwell superficial testing (e.g., 30 N and 30 T scales).
High Rockwell hardness numbers represent hard materials and low numbers soft materials.
Contact Person: Mr. Raymond Chung
CCD vision automatic measurement system: including image capture, calibration, image processing, geometric measurement, document annotation, album management, and fixed-scale printing
Automatic measurement: automatically capture the indentation and measure the indentation diameter and calculate the corresponding Brinell and Vickers hardness values
Max Specimen Height: Rockwell 175mm, Brinell/Rockwell 115mm
Automatic calibration: full image calibration on the intersection points of the calibration grid