Payment & Shipping Terms:
|10x And 20x
|0 - 60 Seconds, Adjustable
|Max Height Specimen 170mm
|Max Horizontal Space 130mm
portable vickers hardness tester,
vickers hardness test
Halogen Light Vickers Hardness Testing Machine with Automatic Turret for Carbonized Layers
Model: iqualitrol iVick-472M
Technical Data Download: iqualitrol Vickers Hardness Tester iVick-472M
Operation Manual Download: iqualitrol Manual of iVick-472M
iqualitrol iVick-472M Motorized Turret Vickers hardness tester adopts load cell control system, improves the precision of the test force and the repeatability and stability of the indicating value.
It can show the test force, dwell time, test numbers on the screen, only need to input the diagonal of the indentation when operation, and can automatically get the hardness value shows on the screen.
Available to install optional CCD image automatic measuring system.
Suitable for ferrous metal, non-ferrous metals, IC thin sections, coatings, ply-metals; glass, ceramics, agate, precious stones, thin plastic sections etc.; hardness testing such as that on the depth and the trapezium of the carbonized layers and quench hardened layers.
|0.3kgf, 0.5kgf, 1kgf, 2kgf, 2.5kgf, 3kgf, 5kgf, 10kgf
|2.94N, 4.90N, 9.80N, 19.6N, 24.5N, 29.4N, 49N, 98N
|10x (Observe), 20x (Measure)
|10x Reading Microscope
|X-Y Test Table
|Size: 100×100mm; Travel: 25×25mm; Resolution: 0.01mm
|Max. Height of Specimen
|ISO 6507, ASTM E92, JIS Z2244, GB/T 4340.2
|530×280×630mm, Packing Dimension: 620×450×740mm
|Net Weight: 35kg, Gross Weight: 47kg
|Instrument Main Body
|10× Reading Microscope
|Diamond Vickers Indenter
|each 1 pc
|Cross Test Table
|Horizontal Regulating Screw
|Halogen Lamp 12V,15~20W
|Inner Hexagon Spanner 2.5mm
|Hardness Block 700~800 HV1
|Hardness Block 700~800 HV10
|Usage Instruction Manual
The Vickers hardness test method consists of indenting the test material with a diamond indenter, in the form of a right pyramid with a square base and an angle of 136 degrees between opposite faces subjected to a load of 1 to 100 kgf. The full load is normally applied for 10 to 15 seconds.
The two diagonals of the indentation left in the surface of the material after removal of the load are measured using a microscope and their average calculated. The area of the sloping surface of the indentation is calculated. The Vickers hardness is the quotient obtained by dividing the kgf load by the square mm area of indentation.
When the mean diagonal of the indentation has been determined the Vickers hardness may be calculated from the formula, but is more convenient to use conversion tables.
The Vickers hardness should be reported like 800 HV/10, which means a Vickers hardness of 800, was obtained using a 10kgf force. Several different loading settings give practically identical hardness numbers on uniform material, which is much better than the arbitrary changing of scale with the other hardness testing methods.
The advantages of the Vickers hardness test are that extremely accurate readings can be taken, and just one type of indenter is used for all types of metals and surface treatments. Although thoroughly adaptable and very precise for testing the softest and hardest of materials, under varying loads, the Vickers machine is a floor standing unit that is more expensive than the Brinell or Rockwell machines.
Contact Person: Mr. Raymond Chung