Payment & Shipping Terms:
|Turret:||Automatic Turret||Objective:||10X, 20X, 40X|
|Height:||Max Specimen Height 185mm||Throat:||Max Horizontal Space 130mm|
|Hardness:||Range 1HV~2967HV||Data Output:||WORD Or EXCEL Report|
portable vickers hardness tester,
vickers hardness test
50Kgf Vickers Hardness Testing Machine with Touch Controller and Internal Vickers Software
Catalogue Download: iqualitrol Vickers Hardness Tester iVick-486ST
New generation of Vickers hardness tester adopts the integrated design of hardness tester and panel computer, all the testing parameters can be set on the panel computer.
CCD image acquisition system can show dynamic indentation image, lock the image and automatically get the Vickers hardness value with high precision and stability, and avoid the human errors.
Touch Screen controller installed with Windows 7 operating system, has all functions of computer and can be connected to the output devices such as external screen and printer;
Original installed three objectives for measurement, automatic recognition and motorized turret.
The lifting screw adopts worm and gear structure with smooth transmission;
Automatic test force loading, dwell and unloading;
Support hardness conversion function;
Automatically save the measuring data, generate the hardness-depth curve and save as WORD or EXCEL document;
With digital micrometer X-Y test anvil, it makes measurement more precise and realizes the fully automation of Vickers hardness testing with optional motorized X-Y test table.
Suitable for ferrous metal, non-ferrous metals, IC thin sections, coatings, ply-metals; glass, ceramics, agate, precious stones, thin plastic sections etc.; hardness testing such as that on the depth and the trapezium of the carbonized layers and quench hardened layers.
0.5kgf, 1kgf, 2kgf, 2.5kgf, 3kgf, 5kgf, 10kgf, 20kgf, 30kgf, 50kgf
|Loading Method||Automatic (Loading/Dwell/Unloading)|
|Computer||CPU: Intel I3, Memory: 2G, SSD: 32G|
|CCD Pixel||1.30 Million|
|Conversion Scale||HV, HK, HRA, HRB, HRC, HRD, HRE, HRF, HRG, HRK, HR15N, HR30N, HR45N, HR15T, HR30T, HR45T, HS, HBW|
|Hardness Reading||Indentation Displaying and Automatic Measuring on Touch Screen|
|Data Output||WORD or EXCEL Report with Curve Chart|
|Objective||10x, 20x, 40x|
|Total Magnification||100x, 200x, 400x|
|Resolution||0.25μm, 0.125μm, 0.0625μm|
|Light Source||Halogen Lamp|
|Motorized X-Y Test Table||Size: 100×100mm; Travel: 25×25mm; Resolution: 0.01mm|
|Max. Height of Specimen||185mm|
|Power Supply||AC220V/50Hz/1Ph, AC110V/60Hz/1Ph|
|Execute Standard||ISO 6507, ASTM E92, JIS Z2244, GB/T 4340.2|
|Dimension||560×335×675mm, Packing Dimension: 650×380×960mm|
|Weight||Net Weight: 48kg, Gross Weight: 62kg|
|Instrument Main Body||1 set||Diamond Vickers Indenter||1 pc|
|10x, 20x, 40x Objective||each 1 pc||Cross Test Table||1 pc|
|Power Cable||1 pc||Horizontal Regulating Screw||4 pcs|
|Level||1 pc||Fuse 1A||2 pcs|
|Halogen Lamp 12V/15~20W||1 pc||Screw Driver||1 pc|
|Inner Hexagon Spanner 2.5mm||1 pc||Hardness Block 700~800 HV1||1 pc|
|Hardness Block 400~500 HV5||1 pc||Anti-dust Cover||1 pc|
|Usage Instruction Manual||1 copy|
Principle of Vickers Test
The Vickers hardness test method consists of indenting the test material with a diamond indenter, in the form of a right pyramid with a square base and an angle of 136 degrees between opposite faces subjected to a load of 1 to 100 kgf. The full load is normally applied for 10 to 15 seconds.
The two diagonals of the indentation left in the surface of the material after removal of the load are measured using a microscope and their average calculated. The area of the sloping surface of the indentation is calculated. The Vickers hardness is the quotient obtained by dividing the kgf load by the square mm area of indentation.
When the mean diagonal of the indentation has been determined the Vickers hardness may be calculated from the formula, but is more convenient to use conversion tables.
The Vickers hardness should be reported like 800 HV/10, which means a Vickers hardness of 800, was obtained using a 10kgf force. Several different loading settings give practically identical hardness numbers on uniform material, which is much better than the arbitrary changing of scale with the other hardness testing methods.
The advantages of the Vickers hardness test are that extremely accurate readings can be taken, and just one type of indenter is used for all types of metals and surface treatments. Although thoroughly adaptable and very precise for testing the softest and hardest of materials, under varying loads, the Vickers machine is a floor standing unit that is more expensive than the Brinell or Rockwell machines.
Contact Person: Mr. Raymond Chung