Payment & Shipping Terms:
|Diameter:||Mould Dia 30mm||Time:||Mounting Time Less Than 10 Min|
|Temperature:||Heating Range 100℃ - 200℃||Pressure:||0-2Mpa|
|Cooling:||Water Cooling||IPress-885:||Mould Dia 45mm|
mounting press machine,
metallurgical sample mounting press
Touch Screen Auto Metallographical Specimen Hot Mounting Press with Mould Dia 30mm
Fast mounting press to improve productive capacity. Optimized heating and cooling steps gives a very short work cycle, usually 1.25inch (30mm) mounted specimen in 6 minutes.
iPress-883 automatic inlaying machine with the function of water cooling, suitable for hot Mosaic of thermosetting material, after set the heating temperature, dwell time, the machine can automatically finish the rest process.
Mounting of specimens is usually necessary to allow them to be handled easily. It also minimizes the amount of damage likely to be caused to the specimen itself.
The mounting material used should not influence the specimen as a result of chemical reaction or mechanical stresses. It should adhere well to the specimen, and if the specimen is to be electropolished later in the preparation then the mounting material should also be electrically conducting.
Specimens can be hot mounted (about 150 °C) using a mounting press either in a thermosetting plastic, e.g. phenolic resin, or a thermosoftening plastic e.g. acrylic resin. If hot mounting will alter the structure of the specimen a cold-setting resin can be used, e.g. epoxy, acrylic or polyester resin. Porous materials must be impregnated by resin before mounting or polishing, to prevent grit, polishing media or etchant being trapped in the pores, and to preserve the open structure of the material.
A mounted specimen usually has a thickness of about half its diameter, to prevent rocking during grinding and polishing. The edges of the mounted specimen should also be rounded to minimize the damage to grinding and polishing discs.
|Sample diameter||30mm, (Opt. 22mm,45mm)|
|Cooling Temperature range||20-90℃|
|Dwell Time Range||0-999s|
|Cooling Mode||Water Cooling|
Contact Person: Mr. Raymond Chung