Payment & Shipping Terms:
|Cutting Section:||85 X 85mm||Rotation Speed:||2800rpm|
|Cutting Wheel:||300 X 1.5 X 32mm||Motor:||3kW, 380V/50Hz/3Ph|
|Cooling:||Water Cooling||Fixture:||Quick Clamping Mechanism|
precision cutting machine,
abrasive cutting machine
Sample Metallographic Cutting Machine Max Cutting Diameter 85mm Manual Cutting Machine
iqualitrol iCut-407 Metallographic Sample Cutting Machine can be used to cut general all-facies and lithofacies sample materials, which could guarantee the cutting of optimum samples under the safe conditions.
The machine adopts all-sealing structure and quick clamping mechanism, and also attaching the cooling water tank for the use of coolant circulation, prolonging the service life, perfecting and facilitate the operation.
The inner and outer covering shell, tongs mouth, operating board, and other main spares are all made of stainless steel material, which is one of the best equipment for sample cutting.
|Max Cutting Section||85 x 85mm|
|Cutting Wheel||300 x 1.5 x 32mm|
|Cutter Dimension||730 x 700 x 540mm|
|Stand Dimension||750 x 730 x 1200mm|
|Weight of Cutter||130Kg|
|Weight of Stand||180Kg|
When cutting a specimen from a larger piece of material, care must be taken to ensure that it is representative of the features found in the larger sample, or that it contains all the information required to investigate a feature of interest.
One problem is that preparation of the specimen may change the microstructure of the material, for example through heating, chemical attack, or mechanical damage. The amount of damage depends on the method by which the specimen is cut and the material itself.
Cutting with abrasives may cause a high amount of damage, while the use of a low-speed diamond saw can lessen the problems. There are many different cutting methods, although some are used only for specific specimen types. iqualitrol provides the iCut-864 low speed diamond saw for cutting OM (optical microscope), SEM (scanning electron microscope), and even TEM (transmission electron microscope) specimen.
Contact Person: Mr. Raymond Chung