Payment & Shipping Terms:
|Loading:||Motorized Loading Control||Reading:||Dial Gage, 0.5HR Resolution|
|Dwell Time:||5 - 60 Seconds, Adjustable||Height:||Max Height Specimen 175mm|
|Throat:||Max 165mm||Standard:||ISO 6508, ASTM E-18|
digital rockwell hardness tester,
hardness testing instruments
Dial Gauge Superficial Rockwell Hardness Testing Machine With Adjustable Dwell Time
iRock-434M Motorized superficial Rockwell hardness tester adopts the mechanism of automatic loading and unloading of the test force, the test force is regulated by the load-change hand wheel.
Dwell time can be preset, except setting zero to dial, there is no man-made error. The instrument has high sensitivity and stability.
Suitable for surface quenched steel, surface heat treating and chemical treating materials, copper alloy, aluminum alloy, sheet, zinc layers, chrome layers, tin layers, bearing steel and cold and hard casting etc.
|Initial Test Force||3kgf (29.42N)|
|Total Test Force||15kgf (147.1N), 30kgf (294.2N), 45kgf (441.3N)|
|Indenter||Diamond Rockwell Indenter, ф1.588mmBall Indenter|
|Loading Method||Automatic (Loading/Dwell/Unloading)|
|Max. Height of Specimen||175mm|
|Execute Standard||ISO 6508, ASTM E-18, JIS Z2245, GB/T 230.2|
|Dimension||520×215×700mm, Packing Dimension: 650×370×950mm|
|Weight||Net Weight: 73kg, Gross Weight: 100kg|
|Instrument Main Body||1 set||Diamond Rockwell Indenter||1 pc|
|ф1.588mmBall Indenter||1 pc||Middle Plane Test Table||1 pc|
|Large Plane Test Table||1 pc||V-shaped Test Table||1 pc|
|Hardness Block 70~85 HR30T||1 pc||Hardness Block 80~90 HR15N||1 pc|
|Hardness Block 65~80 HR30N||1 pc||Weight A, B, C||3 pcs|
|Power Cable||1 pc||Fuse 2A||2 pcs|
|Level||1 pc||Horizontal Regulating Screw||4 pcs|
|Screw Driver||1 pc||Spanner||1 pc|
|Anti-dust Cover||1 pc||Usage Instruction Manual||1 copy|
The Rockwell test consists of measuring the additional depth to which a carbide ball or diamond penetrator is forced by a heavy (major) load beyond the depth of a previously applied light (minor) load (SET point).
The minor load is applied first and a SET position is established on the dial gauge or displacement sensor of the Rockwell tester. Then the major load is applied. Without moving the piece being tested, the major load is removed and, with the minor load still applied, the Rockwell hardness number is automatically indicated on the dial gauge or digital display.
The diamond penetrator is used for testing materials such as hardened steels and cemented carbides. The carbide ball penetrators, available with 1/16 inch, 1/8 inch, 1/4 inch, and 1/2 inch diameter, are used when testing materials such as steel-copper alloys, aluminum and plastics to name a few.
Rockwell testing falls into two categories: Regular Rockwell testing (e.g., C and B scales) and Rockwell superficial testing (e.g., 30 N and 30 T scales).
High Rockwell hardness numbers represent hard materials and low numbers soft materials.
Contact Person: Mr. Raymond Chung