Payment & Shipping Terms:
|Memory:||To Save 400 Results||Data Transfer:||By RS232|
digital rockwell hardness tester,
hardness testing instruments
Tube Inner Wall Rockwell Superficial Hardness Test Machine With LCD Digital Display
iRock-45N performs Superficial Rockwell hardness tests in accordance with established specifications, including BS EN ISO 6508:1999 and ASTM E18.
It comes with a diamond cone indenter and a 1/16” diameter ball indenter, as standard. The machine is provided with a LED display. It is used to display measured hardness results, to monitor the progress of the test and to display derived statistical data.
|Product Name||Superficial Rockwell Hardness Tester|
|Total test force||15Kgf(147.1N), 30Kgf(294.3N), 45Kgf(441.3N)|
|Superficial Rockwell Scales||15T,30T,45T;15W,30W,45W;15X,30X,45X;15Y,30Y,45Y;15N,30N,45N|
|Memory||Max 400 items of test results stored automatically|
|Dwell Time||Adjustable 2-50s|
|Hardness Indication||High definition backlight LCD|
|Loading Control||Auto Loading/Dwell/Unloading|
|Hardness Conversion||HRC, HRB, HRA, HV, HK, HBW, etc|
|Max. Height Of Specimen||260mm|
|Dimension||710 x210 x830mm|
|Power supply||AC 220V/50Hz;AC110V/60Hz|
|Execution Standard||GB/T230.2, JJS Z2245, EN-ISO 6508 , ASTM E-18|
● Horizontal Protrudent Indenter Design, suitable for internal and external testing;
● Testing surfaces difficult to reach by normal Rockwell testers, internal surface of rings and tubes with diameters over 23mm;
● Testing force was set automatically according to testing scale;
● Measuring results displays on LCD and can be printed, or transmitted to external computer automatically by USB or RS-232 data output port;
● Test value can be transformed to the value of HB, HV, HK and σb value;
● Conform standards ISO 6508-2, ASTM E18.
The Rockwell test consists of measuring the additional depth to which a carbide ball or diamond penetrator is forced by a heavy (major) load beyond the depth of a previously applied light (minor) load (SET point).
The minor load is applied first and a SET position is established on the dial gauge or displacement sensor of the Rockwell tester. Then the major load is applied. Without moving the piece being tested, the major load is removed and, with the minor load still applied, the Rockwell hardness number is automatically indicated on the dial gauge or digital display.
The diamond penetrator is used for testing materials such as hardened steels and cemented carbides. The carbide ball penetrators, available with 1/16 inch, 1/8 inch, 1/4 inch, and 1/2 inch diameter, are used when testing materials such as steel-copper alloys, aluminum and plastics to name a few.
Rockwell testing falls into two categories: Regular Rockwell testing (e.g., C and B scales) and Rockwell superficial testing (e.g., 30 N and 30 T scales).
High Rockwell hardness numbers represent hard materials and low numbers soft materials.
|Instrument Main Body||Ø1.588mm Ball Indenter||Weight 1|
|Cone diamond Indenter||Ø1.588mm Steel Ball||Weight2|
|Test block HR30N||Screwdriver for indenter mounting||Weight 3|
|Test block HR30T||Power Cable||Weight 4|
|Accessories Box||Ø 40mm V-shape Anvil||Anti-dust Cover|
|Warranty card||Ø 60mm Flat Anvil||Instrument Manual|
Contact Person: Mr. Raymond Chung
CCD vision automatic measurement system: including image capture, calibration, image processing, geometric measurement, document annotation, album management, and fixed-scale printing
Automatic measurement: automatically capture the indentation and measure the indentation diameter and calculate the corresponding Brinell and Vickers hardness values
Max Specimen Height: Rockwell 175mm, Brinell/Rockwell 115mm
Automatic calibration: full image calibration on the intersection points of the calibration grid